Mini Series Pressure Transducer
With a diameter of only 0.5 inches and an overall length of less than 1.0 inch, the mini series pressure transducer was designed for applications involving a limited amount of space. In addition, this series was also designed to be not only accurate, but rugged as well. With a standard operating temperature of 300º F, optional temperature limits up to 500º are available.
Options: • Range: 01,000 psi to 020,000 psi • LED display • Differential pressure measurement
Specifications: • Max excitation voltage: 10Vdc (5 Vdc for optimum stability) • Full scale output (FSO): 23 mV/V standard • Overload: 50% over rated capacity without zero shift • Nonlinearity: less than 0.3% FSO* • Hysteresis: less than 0.2% FSO • Nonrepeatability: better than 0.05% FSO • Thermal zero shift: less than 0.005% per º F • Thermal span shift: less than 0.010% per º F
* BFSL

SPG/DPG Series Pressure Transducer
The SPG/DPG pressure transducers are our most versatile line of pressure transducers. A number of pressure port options as well as electrical connection options are available. The unit shown to the right has flying leads as its electrical connection and a 9/16" high pressure fitting as its pressure port. Please contact us to discuss your requirements.
Options: • Range: 0500 psi to 020,000 psi • Differential pressure measurement • Optional 400º F or 500º F service available • LED display
Specifications: • Max excitation voltage: 12V max (1,000Ω); 20V max (3,500Ω) • Full Scale Output (FSO): 23 mV/V standard • Nonlinearity: <0.1% FSO for 10,000 psi and higher; <0.2% pressures below 10, 000 psi* • Hysteresis: less than 0.2% FSO • Nonrepeatability: better than 0.05% FSO • Zero balance: typically within ±2% FSO, available down to ±0.5% • Thermal zero shift: less than 0.001% per º F • Thermal span shift: less than 0.002% per º F
*BFSL 

