ULG Series Load Cells
The Miniature ULG series measures only 1.48" x 1.81" x 2.00", including the MS3102E14S6P connector, and was designed to accommodate applications that have only a limited amount of space. If your needs include an accurate and reliable load cell in a small package, please contact one of our sales representatives to discuss your specific requirements.
Options: • Internal cal resistor • LED display • Range: 0250 lbs to 01,000 lbs
Specifications: • Max excitation voltage: 12 Vdc • Full Scale Output (FSO): 1  3 mV/V ±10% standard, available to ±1% • Overload: 50% over rated capacity without zero shift • Nonlinearity: less than 0.20% FSO* • Hysteresis: less than 0.10% FSO • Nonrepeatability: better than 0.05% FSO • Thermal zero shift: less than 0.001% per º F • Thermal span shift: less than 0.002% per º F
* BFSL
 Designed for measurement of tension and compression, the Standard ULG series load cells provide outstanding performance at an affordable price. Originally designed for diesel engine testing, this load cell has proven itself as extremely rugged, surviving when competitive products have failed.
Options: • Aluminum or stainless steel • Internal cal resistor • Range: 01,000 psi to 05,000 lbs • LED display
Specifications: • Max excitation voltage: 12Vdc • Full Scale Output (FSO): 13 mV/V standard ±10% (available to ±1%) • Overload: 50% over rated capacity without zero shift • Nonlinearity: less than 0.20% FSO • Hysteresis: less than 0.10% FSO • Nonrepeatability: better than 0.05% FSO • Zero balance: typically within ±2% FSO, available down to ±0.5% • Thermal zero shift: less than 0.001% per º F • Thermal span shift: less than 0.002% per º F
*BFSL 
 DTG/DUG Series Load Cells
The DTG/DUG load cells are designed for the measurement of tension only or tension and compression. Standard operating temperature is up to 300º F with optional limits up to 500º F available. Standard capacities are 8,000, 10,000, 16,000, 20,000 and 40,000 lbs. Custom design, engineering and manufacturing are available. Please contact us for details.
Options: • Special ranges, accuracies, and calibrations • LED display
Specifications: • Max excitation voltage: 15Vdc • Full scale output (FSO): 2 mV/V ±10% standard • Overload: 50% over rated capacity without zero shift • Nonlinearity: less than 0.15% FSO • Hysteresis: less than 0.10% FSO max typically 0.05% or better • Nonrepeatability: better than 0.05% FSO • Bridge resistance: 1000 ohms nominal • Thermal effect on zero: less than 0.002% per º F • Thermal effect on span: less than 0.002% per º F
*BFSL 

